The AirSense™ 10 AutoSet for Her CPAP Machine with HumidAir™ Heated Humidifier is designed with women in mind. From the ivy leaf design to its special algorithm, this machine is a woman's dream. This is a auto-adjusting CPAP machine that auto-adjusts to the optimum pressure on a breath by breath basis.
For Her Algorithm
One of the most innovative features of the AirSense™ 10 AutoSet for Her is the algorithm developed for women and the manner in which OSA affects them. A standard algorithm responds after 3 flow limited breaths and will move up by as much as 3 cmwp. The For Her algorithm responds after just 1 flow limited breath and will move up by as much as 2.5 cmwp. If needed the algorithm setting can be changed to the standard AutoSetAutoSet™ algorithm version.
Also available from ResMed is the AirSense™ 10 AutoSet CPAP Machine with HumidAir™ Heated Humidifier. It offers only the standard AutoSet™ for auto-adjusting the delivered pressure in a black machine. All other features are the same as the For Her version.
AutoSet Response Available in Soft or Standard
Some people are more bothered by drastic changes in pressure. The AirSense 10 AutoSet for Her comes with two algorithm settings: "soft" or "standard". "Soft" response mode is intended to smooth out the changes in pressure, making it feel a lot less drastic.
HumidAir™ Heated Humidifier
The AirSense™ 10 AutoAdjust for Her CPAP Machine has an internal heating plate. The water chamber is attached directly to the machine for a minimal footprint. The humidifier can be set to 9 different settings: off, and 1 through 8 with 1 being the lowest level of humidification, and 8 being the highest.
Enjoy enhanced humidification by adding the OPTIONAL ClimateLineAir integrated heated tube. The Auto Climate Control mode will consider the room temperature and adjust the level of humidification delivered to the mask.
For added convenience the AirSense™ 10 AutoSet for Her CPAP features SmartStart™ technology. Simply put on the mask and start breathing and the machine will begin therapy. Similarly, when the mask is taken off, the machine will register the machine is no longer in use and turn off.
Color LCD Display with Integrated Light Sensor
The LCD display of the AirSense™ 10 AutoSet for Her features vibrant colors to help clearly identify different stages of therapy. A blank circle fills with green as therapy pressure ramps up. Green and red faces indicate proper or poor mask fit and are also used to indicate if the humidifier is working properly.
The machine also features an integrated light sensor which allows the screen to adjust brightness automatically when ambient light changes. If a restroom break is needed in the middle of the night, the screen won't blind with brightness, and in the morning it won't be too dim to read how therapy went. In addition, the screen will automatically turn off after a short time to avoid extra light in a dark room.
Expiratory Pressure Relief (EPR™)
Now a mainstay on ResMed machines, EPR™ is intended to make therapy more comfortable. Breathing out against full prescribed pressure can sometimes be uncomfortable. EPR™ steps in and helps by lowering the pressure by a set amount on exhalation.
Autoset Response is a comfort feature that can be enabled to provide more gentle pressure increases. If the machine detects a pressure increase is needed, the response feature will ease into the increased pressure to help those with lower pressure tolerance.
The AirSense™ 10 AutoSet for Her will display the following advanced therapy data on the machine display through the "More Info" option in the "Sleep Report" menu:
Weight - Machine with Water Chamber: 2.75 lbs (44 oz.)
Dimensions - Machine with Water Chamber: 10.04 in L x 4.57 in W x 5.91 in H
Weight - Standard Chamber: .54 lb
Weight - Machine with Side Cover (Without Water Chamber): Approx. 2.452 lbs
Dimensions - Machine with Side Cover (Without Water Chamber): Approx. 8 in L x 4.57 in W x 5.91 in H
Dimensions - Travel Bag: 13.24 in L x 10 in W x 5 in H
Weight - Travel Bag Alone: 1.18 lb
Weight - Machine with Humidifier; Power Supply and Cord; SlimLine Tube; Travel Bag: Approximately 5.1 lbs
Weight - Machine with Humidifier; Power Supply and Cord; OPTIONALClimateLineAir Tube; Travel Bag: Approximately 5.25 lbs
Sound Level: 26.6 dBA
Water Chamber Capacity: 380 mL
Weight - 90W Power Supply, Cord and Plug: 1.29 lb
Dimensions - 90W Power Supply Only (Not including Cor or Plug): 5 1/2 in x 1 7/8 in x 1 1/4 in
AC Input Range - 90W Power Supply: 100 - 240V, 50 - 60Hz, 1.0 - 1.5A, Class II
DC Output - 90 W Power Supply: 24V --- 3.75A
Typical Power Consumption: 53W (57VA)
Peak Power Consumption: 104W (108VA)
Operating Pressure: 4 - 20 cm H2O (0.2 cm increments)
Length - SlimLine Tubing: 6 feet
Materials - SlimLine Tubing: Flexible Plastic
Materials - Standard Water Chamber: Injection Molded Plastic, Aluminum, Silicone Seal
Materials - Dishwasher Safe Water Chamber: Injection Molded Plastic, Stainless Steel, Silicone Seal
Materials - Standard Filter: Polyester non-woven fiber
Temperature Range - Optional ClimateLineAir™ Heated Tube: 60 - 80° F (16 - 30° C)
Operating Temperature: 41° F - 95° F (5° C - 35° C)
Storage and Transport Temperature: -4° F - 140° F (-20° C - 60° C)
Operating Humidity: 10% - 95% Relative Humidity, non-condensing
Storage and Transport Humidity: 5% - 95% relative Humidity, non-condensing
Operating Altitude: Sea Level - 8500 Feet (2,591 M); Air pressure range 1013 hPa - 738 hPa
Maximum Temperature - Heating Plate: 154° F (68° C)
Maximum Temperature - Cut Out: 165° F (74° C)
Maximum Gas Temperature: Less than or Equal to 106° F (? 41° C)
IEC 60601-1:2005 Classification: Class II (Double Insulation), Type BF, Ingress protection IP22
Machine Warranty: 2 Years
Patient Weight: Greater than 66 lbs
The chamber should be taken apart and hand washed weekly using warm water and a mild detergent. It should be rinsed well and allowed to air dry out of direct sunlight before reassembling. The chamber should not be washed in a dishwasher or washing machine. The SlimLine hose should be washed in the same manner and allowed to completely dry out of direct sunlight before reattaching to the machine and mask.
Required Settings: This machine does not have required settings.
Unless directed otherwise, CPAP.com will leave this machine at its default settings of: